Physio-Control Lifepak CR Plus Features
The Physio-Control Lifepak CR Plus features the same advanced technology trusted by emergency medical professionals, yet still simple to use. The fully-automatic Lifepak CR Plus AED is designed specifically for the first person to respond to a victim of sudden cardiac arrest. Unlike AEDs with complex prompts and limited energy for defibrillation, the Physio-Control CR Plus combines an easy two-step operation, the right level of guidance, and the capability to escalate to 360 joules when needed.
4.5 lbs fully loaded.
Increases up to 360 joules if needed.
Low cost of ownership.
Simple transition to EMS teams who also use Lifepak products.
Safe and effective.
Physio-Control Lifepak CR Plus Specifications
Height: 4.2” (10.7 cm)
Width: 8” (20.3 cm)
Depth: 9.5” (24.1 cm)
Weight: 4.5 lbs (2 kg)
Waveform: Biphasic truncated exponential, with voltage and current duration compensation for patient impedance.
Output Energy Sequence: Multiple levels, configurable from 150 joules to 360 joules (200 joules min for Japan). Factory default settings of 200J, 300J, 360J.
Output Energy Accuracy: ±10% into 50 ohms, ±15% into 25 to 100 ohms.
Shock Advisory System: An ECG analysis system that advises whether a shock is appropriate; meets rhythm recognition criteria specified in DF39.
The device charges for shock only when the Shock Advisory System advises defibrillation.
Typical: Thirty (30) full discharges or 210 minutes of “on time” with a fully charged device.
Minimum: Twenty (20) full discharges or 140 minutes of “on time” with a fully charged device.
Shock Charge Time: Charge times with a fully charged device: 200 joules in less than 9 seconds, 360 joules in less than 15 seconds.
System Recharge Times: Recharge times with a fully discharged device: Able to deliver 6 shocks or provide 42 minutes of operating time after 24 hours of recharge time and 20 shocks or 140 minutes of operating time after 72 hours of recharge time with a new CHARGEPAK at temperatures above 15° C (59° F).
Lid Release/ON-OFF—Controls device power.
SHOCK button (semi-automatic version)—delivers defibrillation energy. After electrodes are attached to a patient, the fully automatic version of the device delivers a shock, if appropriate, not requiring operator intervention.
User Interface: The user interface includes voice prompts, audible tones and graphic prompts.
Readiness Display: The readiness display shows the device status.
OK Indicator: Shows “OK” when the last self-test was completed successfully. When the “OK” indicator is visible, all other indicators are not visible. The “OK” indicator is not displayed during device operation.
CHARGE-PAK Indicator: When displayed, replace the CHARGE-PAK battery charger.
Attention Indicator: When first displayed, at least six (6) discharges or 42 minutes of operating time remain.
Service Indicator: Service required when displayed.
Energy Sequence: Energy sequence is set to 200J, 300J, 360J.
Motion Detection: The motion detection system is set to on during analysis.
Energy Protocol: The energy protocol is set to increase energy only after a lower energy shock was unsuccessful.
Stack Shocks: Stack shocks option is set to off. Turn-On Prompt: The turn-on prompt is set to provide voice prompts upon power on.
CPR Time: The CPR Time is set to 120 seconds.
Voice Prompt Volume: The voice prompt volume is set to high.
Memory Type: Internal digital memory.
ECG Storage: Dual patient data storage. Minimum 20 minutes of ECG stored for the current patient, summarized data stored for the previous patient.
Continuous ECG – A continuous patient ECG report.
Continuous Summary report – A summary of critical resuscitation events and ECG waveform segments associated with these events.
Event Log report – A report of time-stamped markers, which reflect operator and device activity.
Test Log report – A device self-test activity report.
Capacity: Minimum 200 time-stamped event log markers.
Communications: Wireless transfer to a personal computer.
Data Review: Physio-Control provides an array of tools to meet customer needs for data viewing and analysis.
Operating Temperature: 0° to +50°C (+32° to +122°F).
Long-term storage temperature: +15° to +35°C (59° to 95°F)
Short-term storage temperature: -40° to +70°C (-40° to 158°F) for a maximum of one week.
Atmospheric Pressure: 760 mmHg to 429 mmHg, 0 to 15,000 feet above sea level.
Relative Humidity: 5 to 95% (non-condensing).
Water Resistance: IEC60529/EN60529 IPX4 “Splash proof” with electrodes connected, CHARGE-PAK installed.