LTV 1200 Ventilator
The LTV 1200 is a portable ventilator. LTV Ventilator can be used in-home, intra-hospital transport, or in emergency infield use. The System comes with preset buttons for infants, pediatrics, and adult patients and can be customized per each patient’s needs. It offers both invasive and non-invasive modes of ventilation in Pressure Control, Pressure Support, Volume Control, and Spontaneous Breath Types. With the Vyaire Turbine technology, the LTV 1200 eliminates the need for a high-pressure air source.
Quick Patient Ventilation – Choose infant, pediatric, or adult to quickly configure initial ventilation.
More versatile than transport ventilators and significantly smaller than ICU systems.
The O2 Conserve feature reduces O2 consumption when the time and place of care require it.
The internal battery allows transfer from one power sources to the next without interruption.
LTV 1200 Ventilator
Height: 3.25 In (8.4 cm)
Width: 10.5 In (27 cm)
Depth: 13.5 In (38 cm)
Weight: 14.5 lbs (6.5 kg)
Ventilator input 11–15 VDC
AC adapter input 100–250 VAC
High pressure O2 source 40–80 PSIG (2.8–5.5 bar)
Low pressure O2 source < 80 LPM, < 10 PSIG Controls Ventilation Modes: Control, Assist/Control, SIMV/CPAP and NPPV Breath Types: VC, PC, PS and spontaneous Breath Rate: 0–80 BPM Tidal Volume: 50–2,000 mL Inspiratory Time: 0.3–9.9 sec PC/PS/Spontaneous Flow: 160 LPM Pressure Control: 1–99 cmH2O Pressure Support: 1–60 cmH2O Sensitivity: Off, 1–9 LPM O2%: 21–100% O2 flush: 1–3 min Low pressure O2: On/Off PEEP/CPAP: 0–20 cmH2O Insp/Exp hold: 6 seconds maximum Manual breath: 1x current settings Monitors and Controls Peak Inspiratory Pressure: 0–120 cmH2O Mean Airway Pressure: 0–99 cmH2O PEEP: 0–99 cmH2O Breath Rate: 0–250 BPM Airway Pressure Display: -10–108 cmH2O Exhaled Tidal Volume: 0–4,000 mL Exhaled Minute Volume: 0–99.9 L I:E Ratio: 99:1–1:99 Calculated Peak Flow: 10–100 LPM AutoPEEP: 0–100 cmH2O Static Compliance: 1–999 mL/cmH2O Patient Effort: Green LED Extended Features SBT Ventilator presets (infant, pediatric and adult) Variable rise time Variable flow termination Variable time termination Pressure control flow termination Leak compensation O2 conserve O2 cylinder duration calculation O2 flush Apnea interval Fixed Alarms Disconnect/Sense External power low and lost High and low O2 inlet pressure Internal battery low and empty Ventilator inoperative